Premium level indicator for tank provider? There is AC interference and the voltage is high. For example, for the radar level meter used in the production line, the power supply requirement is 24VDC (typical value), but in the on-site measurement, it is found that the power supply is displayed as 27.2V, which is significantly higher than 24VDC, resulting in a large measurement result and even a radar level meter. crash phenomenon. The installation position of the radar level meter is incorrect, which leads to deviations in the measurement. For example, the accumulation of aggregates in the transfer bin is a “mountain”-shaped cone, but only one radar level meter is installed near the discharge port of the return belt. , the installation position is too close to the discharge opening of the return belt, and at the same time, it is too far from the discharge opening of the feeding belt on both sides. Just below the radar level meter is the drop point of the return belt. If the distance is too close, the aggregate in the falling process will interfere with the radar level meter and form false reflections. Find additional info at level indicator for tank.
In the process of cement production, material level detection and material level alarm control play an important role from mining and crushing to storage and packing of finished cement. But because of the complex working conditions of cement plant, radar level meter has no small difficulty in measuring. Today, I would like to talk with you about specific difficulties and solutions. First cement plant material for measurement in the dust under the condition of particularly large, especially some pneumatic feed bin, involves the kinds of materials, material particle size is differ, dielectric constant is different also, such as broken limestone, raw meal, clinker, fly ash, coal, cement, the level of the bunker, it is very difficult to accurately measure Especially fly ash bin with low dielectric constant.
If the radar level meter is unreasonably selected, the interference echo cannot be handled well, and the reliability of the instrument will be reduced. Therefore, the following factors should be considered when selecting a radar level meter: Conductivity and dielectric constant of the measured medium. The measured medium is a conductive liquid or a liquid with a dielectric constant above 4. Generally, a common radar is selected. Liquids with small dielectric constants (dielectric constants below 2) and some conductive solids often use precision radars or guided wave radars due to the large amount of interference echoes.
The radar level gauge works by electromagnetic waves. Its working principle is to measure the specific liquid level by transmitting electromagnetic waves to the measured target. After the electromagnetic waves are emitted, they are reflected by the medium. For the radar level gauge, its key function is to ensure that it can transmit magnetrol guided wave radar signals smoothly. In our industrial production site, interference often occurs, so which interference sources will affect the measurement of the radar level gauge? let’s see.
Under many operating conditions, ultrasonic level meter and radar level meter are commonly used. Some users are very entangled in the choice of these two level meter and do not know how to choose. Today, let’s talk about the principles and selection principles of these two types of level meter . Principle and selection principle of ultrasonic level meter. Working principle: The ultrasonic pulse probe emits a beam of ultrasonic pulses to the measured medium, and the sound wave is reflected by the liquid surface. The distance between the liquid levels is measured by measuring the time difference between the emission and reflection of the sound waves. Since the ultrasonic level gauge is not a liquid, it can be used to measure corrosive, non-volatile and non-foaming places.
Measuring principle of radar water level meter: The radar level gauge adopts the working mode of transmitting-reflecting-receiving. The electromagnetic wave emitted by the antenna is reflected by the surface of the measured object and then received by the antenna. After measuring the distance from the water surface to the radar antenna, the elevation of the water surface can be calculated according to the elevation of the radar antenna. The radar water level meter adopts pulse wave technology with low power consumption. It can be powered by two-wire 24 VDC, 485 interface output, or can be directly powered by 12 VDC, SDI-12 interface output, with high accuracy and wider application range.
KAIDI level transmitter manufacturer is dedicated in providing complete customized solutions for a wide range of industrial automation process applications – in material level, liquid flow, pressure and temperature. We are constantly developing and innovating, our core vision – “to provide solutions that exceed customers’ expectations. In 2012, the company successfully expanded its operations both locally and internationally, achieving global success and recognition for quality fork type level switch, magnetic level gauge products and services. See extra information on https://www.kaidi86.com/. Suitable for chemical industry, petroleum industry, metallurgical industry, water conservancy and electronic industry, etc.